Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Anyway I now that I’m back, here’s what I managed so far:. Message 1 of 4. It will wait for one operation to stop, before the next one starts. Message 3 of I used the Keithley example with the the driver to make the program.
|Date Added:||13 May 2015|
|File Size:||53.73 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Instrument Control (GPIB, Serial, VISA, IVI)
The sourcemeter is connected via GPIB cable. Message keithleh of 3. Thats why i need interface. I have one gpib cable. On my computer – C: If you need assistance on changing the address, please contact service lakeshore.
Keithley A Labview FET Sweep – Discussion Forums – National Instruments
This is a 2 channel meter, and my plan is to sweep one channel from 0 to 5 volts and measure the voltage it is sourcing, while the other channel is swept from mV 26336a mV so no change in voltaage and measure the current coming through. Or you can just do a measurement and see what value you get returned.
Break the problem into smaller steps and when you get stuck come back to this thread and let us know where you are stuck. I would suggest you start small. Message 2 of Message 4 of 5.
The quickest way to find VISA is to go to a vi’s block diagram and press control-space.
I am setting up a measurement where i need to source a voltage from the keithley and plot the current and resistance. Message 5 of I need to make interface so that i can collect data through simple commands. And what is its unit? There are only two ways to tell kwithley thanks: In general a drive menu is created that is nothing more than a keithlsy block diagram showing all the available functions.
Applications Engineer National Instruments. Can please help me out. Message 7 of Navigating a new driver can be frustrating if keothley are not use to the nuiances of vi drives.
This come out in both the Keithley LCD and in labview Most Active Software Boards: We really can’t help without some specifics. Did you rectify that problem, if so, how did you do it.? The problem I am facing is this. Message 5 of I will measure transport properties while going low temp to high. Perhaps, I was overly sarcastic.
keithley – Discussion Forums – National Instruments
I have attched the IV curve im getting thus. Message 2 of I opened them with labview. Message 2 of 5.